- Blue fluorescence indicates a positive result.
- Same familiar platform as the Colilert 250 test:Ready-to-use reagents, no media preparation.
- Convenient unit-dosed packaging.
- No toxic compounds.
- Definitive results in 24 hours.
- Less than 1 minute of hands-on time for streamlined testing work flow.
- Specifically detects Pseudomonas aeruginosa, reducing the risk of false positives from non-target organisms.
- Results are definitive without additional confirmation steps.
- Simple blue fluorescent change minimises subjective result interpretation.
- Liquid enrichment optimises recovery.
How the Pseudalert 250 test works
The Pseudalert 250 test detects the presence of Pseudomonas aeruginosa in bottled water samples.
The test is based on a bacterial enzyme detection technology that signals the presence of P. aeruginosa through the hydrolysis of a substrate in the Pseudalert 250 reagent. P. aeruginosa cells rapidly grow and reproduce using the rich supply of amino acids, vitamins, and other nutrients present in the Pseudalert 250 reagent. Actively growing strains of P. aeruginosa have an enzyme that cleaves the substrate in the reagent to produce blue fluorescence under ultraviolet light.
The Pseudalert 250 test detects P. aeruginosa at 1 cfu per 250 mL sample within 24 hours.
How to use
Learn how to use the Pseudalert 250 test
Step-by-step guide to using the Pseudalert 250 test
Add reagent to sample (sample will turn yellow), and then incubate at 38°C ± 0.5°C for 24–28 hours.
Frequently asked questions
The Pseudalert 250 test is designed specifically for bottled water samples. Bottled water samples are defined as samples taken from any point in the manufacturing of bottled water (from the original water to the finished product). The Pseudalert 250 test was specifically formulated and tested to ensure it would detect stressed P. aeruginosa in these samples.
The Pseudalert 250 test is based on a bacterial enzyme detection method that signals the presence of Pseudomonas aeruginosa through the hydrolysis of an enzyme substrate. P. aeruginosa cells rapidly grow and reproduce using the rich supply of amino acids, vitamins, and other nutrients present in the Pseudalert reagent. Actively growing strains of P. aeruginosa have an enzyme that cleaves the substrate in the reagent to produce blue fluorescence under ultraviolet light. The Pseudalert 250 test detects P. aeruginosa at 1 cfu per 250 mL sample within 24 hours.
There is no comparator for the Pseudalert 250 test. For comparison, use a negative control when interpreting results.
Yes, if the results are positive. An inoculated Pseudalert test presence/absence sample that is positive for Pseudomonas aeruginosa before 24 hours is a confirmed positive test for P. aeruginosa. However, for an accurate quantitative determination of P. aeruginosa in a sample, the results should be read at 24–28 hours of incubation.
Yes, if the results are negative. If an inoculated Pseudalert test sample is inadvertently incubated for over 28 hours, lack of blue fluorescence is a valid negative test result. However, blue fluorescence that develops after 28 hours is not a valid positive test result; the test should be repeated or verified.
For presence/absence testing, use a sterile, transparent, non-fluorescing vessel. For the Quanti-Tray and Quanti-Tray/2000 systems, any sterile vessel can be used.
Resources & Tools
Tests & Accessories
Search the Water Tests & Accessories tool
IDEXX Water has reference materials and approval documents to support the many products in our water portfolio. Find the document(s) you need by selecting the link below.
Search the Reference & Regulatory Documents tool
Pseudalert 250 test (20-test pack)
Catalog Number: WPSE25020I
Water Customer Support
Australia & New Zealand Head Office
IDEXX Laboratories Pty Ltd.
Unit 6, 38-46
Rydalmere, NSW 2116
Phone (AUS): 1300 44 33 99
Phone (NZ): 0800 838 522
Fax (AU): 1800 634 409
Fax: (NZ): 0800 448 443